|Concept:||The difference in degrees between the abutment face of the oar and the blade.|
|Sizes:||Measured in degrees. The default pitch of the blade is specified by the supplier. This is preferable 0°. Some Concept 2 oars, need an pitch of -3° to be easily interchangeable with other oars.|
|Measure:||Using a measurement setup and inclino/angle/pitch meter. The oar is placed on two blocks of wood. One under the blade - at 6cm from the edge - and one under the sleeve. The block of wood under the blade should be perpendicular to the oar shaft. Calibrate the meter by placing it on the wooden block under the blade and calibrate in the direction you are going to measure. Set the digital meter to zero. Rotate the spirit level of the analog meter so that it is exactly between the two lines. The meter is now calibrated. The oar angle can then be measured on the flat part of the oar sleeve. Finally write the oar pitch on the sleeve with a marker, so that it does not have to be measured again.|
|Adjust:||The pitch of the blade can only be set when gluing the sleeve to the oar.|
This article was translated automatically and is provided to you for free. You are most welcome to improve it!